Residual chlorine online analyzer is a multi-parameter instrument suitable for disinfection applications. It can measure pH/ORP
, temperature and flow at the same time. It has three relay outputs, two current outputs that support proportional control, and RS485 serial interface. (Modbus protocol), optional pulse output, supports time display and data recording functions. Residual chlorine online analyzer is a multi-parameter instrument suitable for disinfection applications. It can measure pH/ORP/residual chlorine, temperature and flow at the same time. It has three relay outputs, two current outputs that support proportional control, and RS485 serial interface. (Modbus protocol), optional pulse output, supports time display and data recording functions. Please pay attention to the following points for electrical maintenance of residual chlorine electrodes: 1. The electrodes should be cleaned regularly. The oxygen permeable membrane should not be broken when disassembling and cleaning the electrodes, and the oxygen permeable membrane on the electrode should not be wiped with filter paper, so as not to damage the oxygen permeable membrane. 2. The cable connector must be kept clean and must not be affected by moisture or water. 3. When the displayed value is very different from the actual value, the electrolyte in the electrode may be dry, and the electrolyte needs to be refilled. Generally, the maintenance work of replacing or adding electrolyte is carried out every 6 months; when the permeable membrane is broken, it is necessary to Replace the spare one-piece membrane head. The electrode needs to be repolarized and calibrated every time the electrolyte is replaced or added or the spare integrated membrane head is replaced. The specific steps are as follows: Unscrew the black part of the electrode shield at the lower part of the electrode, take off the spare integrated membrane head, turn the cylinder upside down (don’t let the breathable membrane touch other things), pour the original electrolyte, and in the integrated Fill the membrane head half-full with electrolyte and then put it into the inner core of the oxygen electrode, let the excess electrolyte drain off, and tighten it to ensure that the membrane can be attached to the head of the electrode core. After replacing the electrolyte, the polarization and calibration should be carried out again. When calibrating, a solution of known concentration must be used as a reference standard for calibration. 4. Electrode polarization: After the electrode is connected to the instrument, the electrode is continuously powered on for more than 2 hours, which is polarization, and the calibration can only be carried out after the electrode is polarized. 5. When the water is cut off for a long time on site or the instrument is not used for a long time, the electrodes should be taken out in time, cleaned and put on the protective cap. 6. If the electrode fails, replace the electrode.
To be the best worldwide provider of higher-value water analyzer and the center for quality employment opportunities.
If you already use water analyzer elsewhere or want the ability to offer restricted chat access to certain individuals, water analyzer water quality monitoring device offers you the most flexibility.
To derive the optimal value out of water quality monitoring device water analyzer for your home, make sure they're purchased from a globally certified organization to ensure quality in use. Such an offer can be found at BOQU Water Quality Analyzer
Although there are various available in the market (such as water quality monitoring device, water quality monitoring device, and water quality monitoring device), recent study results have made this water quality monitoring device water analyzer a preferred water analyzer choice of the people.
The best way of water quality monitoring device is to get a water quality monitoring device water analyzer.