Inspection items and related requirements of the online residual chlorine
detector The online residual chlorine detector
has been calibrated by the DPD method before leaving the factory. If the measured value is found to be inaccurate during use, it is necessary to re-zero and calibrate the instrument. 1. For zero adjustment, add about 1 ml of deionized water into the colorimetric tube, then add 1 piece of DPD residual chlorine rate test reagent tablet, soak for 1 minute, crush it with a clean glass rod (accessory) to fully dissolve, and then add Chlorine water to the 10ml mark and shake well, dry the water stains on the glass surface of the colorimetric tube with a roll paper and put it into the colorimetric seat of the instrument, align the scale line on the measuring cell of the instrument and cover it with a light-shielding cover. Quickly adjust the 'zero adjustment' potentiometer on the rear panel of the instrument within 4-5 minutes after adding the DPD reagent, so that the display is 0.00mg/L. 2. Calibration Add about 1 ml of 2.5mg/L--DPD method residual chlorine standard solution into the colorimetric tube, then add a piece of DPD residual chlorine speed test reagent tablet, soak for 1 minute, crush and dissolve with a clean glass rod, Add 2.5mg/L--DPD residual chlorine standard solution to the 10ml scale, dry the water stains on the glass surface of the colorimetric tube with roll paper and put it into the colorimetric seat of the instrument, and align it with the scale on the measuring pool of the instrument line and put on the light-shielding cap. Quickly adjust the 'calibration' potentiometer on the rear panel of the instrument within 4-5 minutes after adding the DPD reagent, so that the display is 2.50 mg/L. Note: When the o-toluidine reagent method is used for measurement, the user should prepare the potassium dichromate residual chlorine standard solution and the use solution by himself, and the instrument can only be used after calibration. For the configuration method and use of the standard solution, please refer to the GB/T Standard Test Method for Drinking Water. The residual chlorine meter is a precision measurement instrument combined with photoelectricity. Before operation, you should read the instructions carefully and operate correctly to obtain accurate measurement results. Requirements for the use of online residual chlorine meter: 1. The use environment must meet the working conditions. 2. The measuring tank must be clean and dry for a long time without dust, and the shading cover must be covered when not in use. 3. If it is used in humid climate, the start-up time must be extended accordingly. 4. The solution to be tested should be poured carefully along the wall of the sample bottle to prevent air bubbles from affecting the measurement accuracy. 5. After replacing the sample bottle or standard solution and after maintenance, it must be re-calibrated. 6. When the word 'LOBAT' of low voltage is displayed on the upper left corner of the display, the battery should be replaced in time to avoid affecting the measurement. 7. After adding the DPD reagent, the whole test should be completed within 4-5 minutes, otherwise the test result may be inaccurate. During the test, the scale line on the colorimetric bottle should be aligned with the scale line on the test seat of the instrument and pressed to the bottom (the interior is positioned by a steel ball). 8. Non-professional maintenance engineers, please do not open the instrument for maintenance.
In the past few decades, water quality monitoring device production has increased because of the use of water analyzer.
Super quality are in offer at BOQU Water Quality Analyzer
, welcome to visit us.
Shanghai Boqu Instrument Co., Ltd. emphasizes our commitment to quality in our laboratory and R&D services.