Water is the most important resource in our life, even more important than food. In the past, people drank raw water directly, but now that people's technology has developed, pollution has become serious, and the water quality has naturally been affected. Some people found that raw water contained a large number of parasites and bacteria, so people used chlorine gas for disinfection, but excessive chlorine content would also cause harm to the human body, so a residual chlorine analyzer
appeared. The residual chlorine analyzer is mainly composed of an electronic unit and a measuring unit (including a flow cell and a residual chlorine sensor), and has the characteristics of calibration-free, maintenance-free, high precision, small size, and low power consumption. The display instrument has the function of slope correction, zero point correction, real-time display of measured value, automatic temperature compensation, and manual compensation of pH value. After the compensation calculation of the electrode signal, it is converted into a more accurate residual chlorine signal. Let's learn about the correct installation method of the residual chlorine analyzer: 1. Installation steps (1) First select the location where the detector is installed, then mark the exact location and drill a hole. (2) On the mounting surface, properly tighten or bolt the detector horizontally. Before installing the detector, be sure to open the upper cover and take the lower cover away. The screws on the cover should be taken away with the cover and they cannot be separated, and the opened upper cover should be dragged aside to hang. (3) After the detector is installed, the upper cover can be closed and tightened, and the lower cover can only be closed after all the wires are connected. (4) Mark the position where the instrument is to be installed and drill holes. At the same time, tighten or fasten the instrument bracket moderately on the punched hole, and clamp it on the bracket. 2. Electrical connection (1) There are two cables between the instrument and the detector. The wires on the instrument have been installed, and the numbers on each cable have been marked to correspond to the numbers of the terminals at the bottom of the detector. Pass the wires through a hole with a diameter of 13mm at the bottom of the detector and at the same time put the wires The glands are placed in the corresponding holes. (2) When the cables have been aligned, tighten the glands, and at the same time do not pull the cables too tightly. (3) If required, connect the external alarm indicator and residual chlorine output program controller to the detection. (4) If the power supply is 240V or 110V, you can directly insert the plug into the socket and clamp it with a spring clip. (5) If the power supply is 24V, then the wire must enter from the re-drilled hole at the bottom of the detector and connect to the 24V terminal. (6) Then put the lid on and tighten it.
The use and installation of water analyzer is compared with most other systems for managing the water quality monitoring device effectively and no doubt water analyzer have won the race so many times.
Shanghai Boqu Instrument Co., Ltd. is a rapidly growing Manufacturing Company based in China. We offer a wide range of equipment that helps in water quality monitoring device which are safe, durable and economical. We provide water analyzer, water quality monitoring device, water quality monitoring device,etc. more about and water analyzer solutions are covered in BOQU Water Quality Analyzer
. Take a visit!
Shanghai Boqu Instrument Co., Ltd. believes that the average profitability will be sufficient.