This is an intelligent digital displayed product for pressure testing and controlling. It integrates functions of measuring,display,output and control all in one.It has a complete electronic structure. Oil-filled piezo-restive pressure sensor with diaphragm is applied in the front part. The output is processed by high-precision & low-temperature drift amplifier,then transformed by high accuracy
A/D converter into digital signal that could be processed by MPU(Micro Processor Unit).
The processed signals control two switches then to test&control the pressure.
With flexible application,simple handling,easy debugging and high reliability,this product is widely used applied to test & control the pressure of fluid medium in many industries including areas of hydroelectricity,city water,oil,chemical,machinery hydraulic system etc.
1. Mechanical connection
It can be directly mounted in the hydraulic pipe by the connecting thread,M20*1.5. In critical application,such as server vibration or shock,soft hose is recommended to use.
2. Electrical connection
To avoid electromagnetic interference,please know
A.The cable system should be as short as possible
B.To use shielded wire
C.To keep way from any interference resources,for example,electric appliances and devices.If installed by soft hose,its body should be grounded independently.
AL1H: Switch connection point for switch 1(when pressure reaches this value,indicator on)
AL1F: Switch connection point for switch 1(when pressure reaches this value,indicator off)
AL2H: Switch connection point for switch 2(when pressure reaches this value,indicator on)
AL2F: Switch connection point for switch 2(when pressure reaches this value,indicator off)
FILT:Filter coefficient. To avoid digit display from fluctuating which is caused by pressure change.after 3-10seconds,it can be set END then save then exit.The bigger the filter coefficient is,
The more stable it is,but the more hysteric.
(How to finish the settings as following)
Set switch point 1 at upper-limit alarm output(normally open status),connect at 4Mpa and disconnection when lower than 3.95Mpa,response delay is 3 seconds
Set switch point 2 at lower-limit alarm output(normally close status),connect at 10Mpa and disconnection when lower than 9.95Mpa,response delay is 10 seconds
Enter the menu:Set
● Press “SET”
● “LOCK”sign(remind you to key in password,0001)
● Press ▲ or ▼ to key in the password
● Press “SET” to confirm
● Press ▲ or ▼as page up or page down to select
(AL1H, AL1F, AL2H, AL2F, END)
● Press “SET” to enter selection menu
● Press ▲or ▼ to alter the settings
● Press “SET” to confirm,or press ▲or▼to enter other menus for other settings
Hello, please leave your name and email here before chat online so that we won't miss your message and contact you smoothly.