Contact us right away
If you run into any issue or have any question.
E-mail: email@example.com | Expert in Water Quality Measurement and Water Treatment Project
Slope cablibration of SD-500P Online Color Meter
1) fill in the Standard buffer in Chamber.(here is 500PCU)
2) Put the cover on.
3) wait around 5 minutes.
4) then move the cursor to "SetSam",
5) then presse the "Enter" after cursor flashes several time.
6) then Slope calibration finished.
Note：please ensure the Cover on Chamber when calibration, becasue the light will have effect.
SD-500P Online Color Analyzer
Resolution:0.1 and 1PCU
Storage time:>3 years(8G)
Recording interval:0-30 Minutes can be setup,
Default 10 minutes
Cleaning method:Manual cleaning
Analog output:4~20mA output
Relay output:Four SPDT,230VAC,5A;
Fault alarm:two Acousto-optic alarm,
Alarm value and time can be set
Power Supply:AC,100~230V,50/60Hz or 24VDC ;power comsumption: 50W
Sample flow rate:0mL~3000mL/min,
Ensure that the flow rate is no bubbles
It will more accuracy in low flow rate for low range measuring
Inflow pipeline:1/4" NPT,(Provide external interface)
Outflow pipeline:1/4" NPT,(Provide external interface)
BOQU Contact Info:
Facotry Address:No. 118 Xiuyan Road,Pudong New Area,Shanghai,zip code:201315,China
Hello, please leave your name and email here before chat online so that we won't miss your message and contact you smoothly.